A. Multi-Product Calibrator. Service Manual. PN August Rev.5 , The Fluke Model A Multi-Product Calibrator (Figure ) is a precise. Fluke Corporation. 1. A Specifications. The following paragraphs detail specifications for the A Calibrator. changes more than 5 °C. (See “Zeroing the Calibrator” in Chapter 4 of the A Operator Manual.) Also see additional. 2 Fluke Corporation A/A Multi-Product Calibrators Extended Specifications. Warmup Time. Twice the time since last warmed up, to a maximum of
|Published (Last):||24 September 2005|
|PDF File Size:||9.67 Mb|
|ePub File Size:||3.90 Mb|
|Price:||Free* [*Free Regsitration Required]|
Edge Rise Time Tabie Verification All of the Oscilloscope Calibration functions should be verified at least once per year, or each time the SC is calibrated. V oltage Mode D Apply power sensor fluke 5500a calibrator manual factor mwnual 1 0 MHz W: Table shows the test points for frequency. By pressing ‘print’ button you will print only current page. The actual aclibrator and capacitance values are entered while they are being measure by the Calibrator Mainframe.
How fluke 5500a calibrator manual Contact Fluke Note This procedure should only be used for adjusting the leveled sine wave harmonics.
The A sources specific output values and you measure the outputs using traceable measuring instruments of higher accuracy.
Full text of “FLUKE A Service”
All equipment specified for SC calibration must flhke calibrated, certified traceable if traceability is to be maintained, and operating within their normal specified operating environment.
The far right of the peak is fixed at the far right of the centerline, as shown below. Specifications for the standard A are provided at the fluke 5500a calibrator manual of this chapter. Fluke 5500a calibrator manual 95 V to V, its output is a square wave-like signal that alternates between the negative peak and the positive peak, with the centeriine at -1 0 V.
Page – Oscilloscope Input Resistance Measuremen Synthesized Capacitance Function If the warning still occurs, repair may be necessary. Adjust the dc offset on the 1 B so the last ns of the fluke 5500a calibrator manual of the square wave is on the center line.
Specifications for the Oscilloscope Option are in Chapter 6. These specifications are valid after allowing a warm-up period of 30 minutes, or twice the time the Cxlibrator has been turned off.
The harmonics need to he adjusted so that the fluke 5500a calibrator manual harmonic is at 40 dBc and third harmonic should typically he at 50 dBc as shown in Figure DC Current Specifications Lift off the guard box cover using the finger pull on the rear edge of the cover. Continue with the high frequency fluke 5500a calibrator manual. Table below summarizes the DMM settings required to make topline and baseline measurements.
As you adjust it, the spur will move down the waveform, towards the right.
Another example, fluke 5500a calibrator manual phase uncertainty of a 60 Hz fundamental output and a Hz harmonic output is fluke 5500a calibrator manual degree.
The 50 O termination is closest to the HP A input. Enter 10 Flluke into the Calibrator Mainframe. The signal is then shaped and split to generate the fast edge and external trigger signals. Multiply the peak-to-peak value by 0. The voltage is applied to the external 50 O load, and the circuit current is monitored by the A6 DDS board.
For this test, set the A output to sinewave. If an out-of-tolerance condition is found, the instrument can he re-calihrated using the front panel or the remote interface as described previously fluke 5500a calibrator manual this chapter.
Repeat step 5 until the Calibrator Mainframe display indicates that the next steps calibrate Leveled Sine flatness. Leveled Sine Wave Eunction Specifications FR A5 interface fault Is the A5 fluke 5500a calibrator manual installed?
Install a copper short circuit in the front panel TC connector total instrument zero only. Generating a Calibration Report Three different calibration reports are available from the A, each one either formatted for printing, or in comma- separated variable format for importation into a spreadsheet. FR A7 Shunt amp fault 2.
Calculating Power Uncertainty Overall uncertainty for power output in watts or VARs is based on the root sum square rss of the individual uncertainties in percent for the selected voltage, current, and power factor parameters: Wave Generator Verification This procedure uses the following equipment: The calihrator meter should fluke 5500a calibrator manual approximately 75 mW.
AC Current Calibration Steps Diagnostics runs the following tests: Wave Generator Verification at 1 1.
Low Frequency Flatness Verification at 5. Leveled Sine Wave Frequency Verification Equipment Setup This procedure uses the spectrum analyzer. Program the Calibrator Mainframe to output as listed in Table The highest compliance voltage Is limited to 2.
Other possible suspects include U15 and U48, all on the A6 assembly. Low Erequency Calibration Adjust R16 so that the wave crosses the horizontal center line one division before the fluke 5500a calibrator manual center.
The A then goes into Operate, or asks you to place it into Operate.